G) MONOPOLAR OUTPUT RECEPTACLE
The monopolar output for the handpiece is on the lower right front of the unit. The handpiece plug is designed to fit in only one direction. Plug the connector from the handpiece into the receptacle on the bottom of the unit. The three button handpiece is designed to give the doctor complete fingertip control of the power output settings.
H) GROUNDING PLATE OUTPUT RECEPTACLE
To use the optional grounding plate with cord (A802EU), insert the plug of the cord into the grounding plate output receptacle (H) and connect the other end into the grounding plate. The plate should be placed underneath the patient at a point where the entire plate is covered by bare skin. The use of conductive gel is recommended.
I) BIPOLAR OUTPUT RECEPTACLE
To use bipolar forceps, you need the following optional accessories, bipolar cord #A827V, footswitch #A803 and a bipolar forceps (please refer to the catalog to choose the right forceps for your particular procedure). Insert the plug into the bipolar outputs receptacle (I), then connect the cord to the forceps.
J) POWER CORD
Plug the female end of the power cord into the base of the unit.
K) FOOTSWITCH OUTPUT RECEPTACLE
The optional footswitch (A803) is plugged into the footswitch output and placed on the floor. The footswitch can be used with monopolar procedures and must be used with bipolar procedures.